4.5mm Cartridge Treatment Guidelines
4.5mm Cartridge Treatment Guidelines is to provide you with a high level of safety consistent with your disease, effective and comfortable treatment. DS-4.5 4MHZ frequency used to transmit high- energy focused ultrasound to the skin, subcutaneous tissue, the ultrasound to penetrate the skin to a depth of 4.5mm, direct subcutaneous SMAS layer, forming “heat coagulation” regional, targeted to thicker skin, such as cheeks, etc.
However, it is very important to know how to correctly use the 4.5mm Cartridge before hifu lifting treatment.
Let’s learn more about 4.5mm treatment guidelines as follows:
4.5mm Treatment Guidelines
When treating the head with hifu machine, please do the treatment according to client’s endurance, and move in each row area as appropriate.
- The implementation of the first line, the energy as close as, possible to the mark line, but do not hit the mandible. Once the image of the mandible appears on the screen, slide the Tranducer slightly back and forth, leaving the jaw away from the positioning indicator on the screen.
- Do not let the center of the energy line than the thyroid cartilage area mark line.
- Treatment is the direct chance of energy firing to the mandible position will increase the potential side effects.
When treating the cheek,drag the patient according to the number of times, and move between the rows according to the situation
- Do not treat the cheekbones or mandible directly.
- To ensure that each row of treatment energy lines are interconnected but not overlapping, and appropriate treatment density.
- Treatment of energy directly to the mandible or to the facial nerve position, will increase the possibility of potential side effects.
Place the Transducer in the outer orbital area, and the end of the power cord toward the eye. In the corner of the mark line up and down the planning of a good energy line.
- When the treatment of the transducer at the corner of the eye mark, make the end of the transducer as the center point. Move the transducer, so that the firing energy presented as fan image.
- Each hit an energy line need to re-positioning Tranducer, can not only turn Tranducer front, so as not to repeat the same location.
Place the Tranducer in the eyebrow area with the end of the power cord facing the eye. Smoothes the planned energy line evenly in the area.
When necessary, adjust the length of the energy line:
– The image sees the edge of the positioning indicator that has fallen on the forehead position.
– Prefrontal bone over the surface has been affected when the fit and energy are fixed.
– When the patient’s pain is noticeable.
The main operation should not let the energy line in any one will hit the eye position, if the green positioning on the screen indicator line in the bone position, do not hit the energy.