3.0mm Cartridge Treatment Guidelines
3.0mm cartridge treatment guidelines of hifu facial is very useful to treat skin tightening of face area. DS-3.0 mm for transmitting high frequency of 8 MHZ focused ultrasound energy into the dermis layer of the skin tissue, the ultrasound to penetrate the skin to a depth of 3.0mm, is responsible for activating the skin’s dermal layer of collagen, effectively enhance the effect of the consolidation of the outline, but also improve large pores and reduce the appearance of wrinkles.
3.0mm Treatment Guidelines
3.0mm – Deep dermis tightening
When treating the head, Involve the Transducer in the outer orbital area according to the patient’s response, and move the end of the power cord toward the eye.
- The implementation of the first line, the energy as close as possible to the mark line, but do not hit the mandible. Once the image of the mandible appears on the screen, slide the Tranducer slightly back and forth, leaving the jaw away from the positioning indicator on the screen.
- Do not let the center of the energy line exceed the marker line of the thyroid cartilage area.
- The energy works directly to the mandible position during the treatment will increase the potential side effects.
When treating the cheek,drag the patient according to the number of times, and move between the rows according to the situation.
- Do not treat the cheekbones or mandible directly.
- To ensure that each row of treatment energy lines are interconnected but not overlapping, and appropriate treatment density.
- Treatment of energy directly to the mandible or to the facial nerve position, will increase the possibility of potential side effects.
Place the Transducer in the outer orbital area, and the end of the power cord toward the eye. In the corner of the mark line up and down the planning of a good energy line.
- When the treatment of the transducer at the corner of the eye mark, make the end of the transducer as the center point. Move the transducer, so that the firing energy presented as fan image.
- Each hit an energy line need to re-positioning Transducer, can not only turn Transducer front, so as not to repeat the same location.
Area above eyebrows
Place the Tranducer in the eyebrow area with the end of the power cord facing the eye. Smoothes the planned energy line evenly in the area.
When necessary, adjust the length of the energy line:
– The image shows the edge of the positioning index has fallen on the forehead position.
– Prefrontal bone over the surface has been affected when the fit and energy fixed point.
– Patient pain response was more significant
Under The Eyes
- The Transducer at the eye under the mark line, while gently down the infraorbital skin.
- Firing the first row of the top of the energy line, and then hit the next one, the process to keep Transducer angle correct, do not tilt.
Basic Operation Requirement
Don’t put the transducer near eye, don’t put the transducer on the bone area (green line notes)